

Electricity Meters Threephase meters Multifunction meters PSCH4TM.05MD Nominal voltage  3x(57,7115)/(100200) or 3x(120230)/(208400) V
Nominal (maximum) current  1(2) or 5(10), 5(80)
Accuracy class  0,5S or 1,0 (1,0 or 2,0)
Number of tariffs  4


Electricity Meters Threephase meters Multifunction meters PSCH4TM.05MN Nominal voltage  3x(57,7115)/(100200) or 3x(120230)/(208400) V
Nominal (maximum) current  1(2) or 5(10), 5(80)
Accuracy class  0,5S or 1,0 (1,0 or 2,0)
Number of tariffs  4


Electricity Meters Threephase meters Multifunction meters PSCH3ÒÌ.05Ì, PSCH3ÒÌ.05D Nominal voltage – 3*(120230)/(208400) V Nominal (maximum) current – 5(100) À Accuracy class when measuring active (reactive) energy – 1,0 (2,0) Number of tariffs  4 Communication interfaces – independent RS485 and optoport which are of equal priority. ModBussimilar, SET4ÒÌ.02 compatible exchange protocol with the extended addressing possibility. PSCH3ÒÌ.05D modification is to be fixed on DINbar. 

Electricity Meters Threephase meters Multifunction meters PSCH4ÒÌ.05D
Nominal voltage –3*(57,7115)/(100200) or 3*(120230)/(208400) V
Nominal (maximum) current – 1 (1,5) or 5 (7,5) À
Accuracy class when measuring active (reactive) energy  0,5S (1,0)
Number of tariffs  4
Communication interfaces – independent and of equal priority: RS485 and optoport
ModBussimilar, SET4ÒÌ.02compatible exchange protocol with the extended addressing possibility.
Electric power meter is designed for installation onto DINrail (type ÒÍ35 in compliance with GOST R MEC 607152003).


Electricity Meters Threephase meters Multifunction meters PSCH4ÒÌ.05Ì Nominal voltage –3*(57,7115)/(100200) or 3*(120230)/(208400) V Nominal (maximum) current – 1 (1,5) or 5 (7,5) À Accuracy class when measuring active (reactive) energy  0,5S (1,0) Number of tariffs  4 Communication interfaces – independent and of equal priority: RS485 and optoport ModBussimilar, SET4ÒÌ.02compatible exchange protocol with the extended addressing possibility. 

Electricity Meters Threephase meters Multifunction meters PSCH4ÒÌ.05ÌK Rated voltage – 3õ(57.7115)/(100200) or 3õ(120230)/(208400)
Rated (maximum) current intensity – 1(2) or 5(10) 5(100) A
Accuracy class when measuring active (reactive) energy – 0.5S or 1.0; 1.0 or 2.0
Number of tariffs  4
Communication interfaces – independent, equally prioritized: RS485 and optical port
ModBussimilar, SET4TM.02compatible exchange protocol with the option of extended addressing
Plugin modules: GSMcommunicator, PLC modems, Ethernet or ISM


Electricity Meters Threephase meters Multifunction meters SET4ÒÌ.03Ì, SET4ÒÌ.02.2Ì Nominal voltage – 3*(57,7115)/(100200) or 3*(120230)/(208400) V Nominal (maximum) current – 1 (2) or 5 (10) À Accuracy class when measuring active (reactive) energy – 0,2 S or 0,5S (0,5 or 1,0) Number of tariffs  8 Communication interfaces – independent and of equal priority: two RS485 and optoport (SET4ÒÌ.03Ì); RS485 and optoport (SET4ÒÌ.02Ì) ModBussimilar, SET4ÒÌ.02 compatible exchange protocol with the extended addressing possibility.





New products
Timeschedule of workshops
Copyright Copyright © 2012 Nizhny Novgorod Factory named after M.V. Frunze. All rights reserved.
